Sito in Italia dove è possibile acquistare la consegna acquisto Viagra a buon mercato e di alta qualità in ogni parte del mondo.
100 mg/dL(5.6 mmol/L) and the coefficient of variation should
YUWELL Blood Glucose Test Strip Instruction
The product is used for monitoring the blood glucose level. It is not used for
diagnosis, screen nor for new born baby screen, metabolic disorder tests.
Expired, damaged or polluted test strips are forbidden to use. Test shall be
performed by strictly following the rules; otherwise, false result might cause.
Lancing device can only be used for single person, exclusively; lancet is one time
Test strip vial should be kept away from child reach.
To discard the used test strip. Please follow the local environmental affairs
The testing result is for monitoring only; do not change any medication
without consulting healthcare professional.
This test strip should be used with Yuwell Blood Glucose Monitoring System and
to monitor glucose concentration for capillary whole blood.
Testing range: Hematocrit (Hct) range: 30%~60%. Hct lower than 30% will cause
a faulty high result; while, Hct higher than 60% will cause a faulty low result.
, or it may lead to wrong test result.
RH <85%, or it may lead to wrong test result.
When blood volume is lower than 0.8ul, it may lead to wrong test result. Please
Special circumstances (such as dehydration, oxygen deficit, hyperglycemia state
of high permeability, low blood pressure, shock and acidosis, etc.) may lead to
glucose is oxidized to gluconic acid and electrons are produced from the reaction.
When receiving oxygen therapy, it may lead to lower blood glucose level.
A potential, generated by glucose meter, is provided to measure the electron flow,
When the substances follow are beyond restriction, it may affect the test result:
electrical current. The magnitude of current is then measured, converted and
When substances like acetyl aminophenol, vitamin c, dopamine, L-dopa,
correlated to the glucose concentration level of the blood sample.
uric acid, etc. of blood is higher than normal condition, it may lead
to higher blood glucose level. When substances like ibuprofen and sulfamoyl in
human body is higher than normal condition
level. When cholesterol in blood is higher than 500 mg/dL(27.8 mmol/L) or TG is
higher than 3000 mg/dL(166.7 mmol/L), it may lead to abnormal results.
on, automatically. Confirm the code shown in the meter's screen with the code
shown on the vial's (or pouch's) label. Please finish the blood glucose testing in 3
Keep in the vial, away from direct sunshine.
The test strip should be stored in the original strip vial and at 4~30
Suck blood sample. When the screen shows flashing " ". Touch the
blood sample at the edge of strip tip and the blood will be sucked, automatically.
The test strip has 24 months shelf-life in sealed condition.
The test result will show in 8 seconds.
Discard any remaining test strips when the vial has been used for 3 month.Take
one piece of test strip and recap the vial, immediately.
Tear the single pouch and use it to test.
Testing result is displayed in unit of mg/dL(mmol/L).
Test strip is for one time use only. Please use it to test as soon as possible.
Testing result range is between 20 mg/dL(1.1 mmol/L) and 600 mg/dL(33.3
mmol/L) and has been rectified to correspond to the result of biochemical
Change the code chip every time when opens a new test strip vial.
Confirm the code shown in the meter's screen with the code shown on the vial's
If the meter shows "LO", It means that the result is lower than 20 mg/dL(1.1
mmol/L), If the meter shows "HI", It means that the result is higher than 600
Leave the code chip in the meter until the new strip test vial and code chip used.
mg/dL(33.3 mmol/L). Please test again. If it shows "LO" or "HI" again, consult
your doctor's instruction for a recheck or treatment.
Diagnostic criteria for diabetes mellitus
Place the meter, test strip, lancing device and a lancet.
Wash hands with warm soapy water and dry them well.
Critera for the Diagnosis of Diabetes mg/dL(mmol/L)
Insert the test strip into the meter. The meter turns on, automatically.
Choose a lancing depth. Generally, finger tip is the suitable position.
Do not lancing in swelling or infected area.
Shoot and collect a tiny blood drop.
Do not squeeze the lanced position.
American Diabetes Association Complete Guide to Diabetes, 4th Ed., ISBN 1-58040-237-2, 2005
75 mg/dL(4.2 mmol/L), the deviation between
monitoring system and lab result should fall within 15 mg/dL(0.83 mmol/L) for the
75 mg/dL(4.2 mmol/L), the deviation between
Insert the test strip. Take a test strip and re-cap it, immediately, to avoid
monitoring system and lab result should fall within
damp. Insert the test strip into the slot with the "YUYUE" side up. The meter turns
TenStep Papel Suplementar 29 de abril de 2005 Prepare-se para o Exame de PMP® (Este documento reflete as experiências do autor Tom Mochal e não é projetado para satisfazer todos os requisitos das pessoas que querem o certificado de Quando eu estava no mercado de trabalho no meio da década de 90, eu não recordo de ver nenhum trabalho onde havia uma exigência ou uma preferên
Chiesadomestica.net Pillola abortiva per mascherare l'orrore dell'aborto 18 novembre 2007 "Meglio l’ospedale di quell’inferno” Ombre e dolore sulla kill pill (e gli effetti collaterali si moltiplicano) - di Assuntina Morresi, Eugenia Roccella Roma. La RU486, la pillola abortiva, sembra avere un merito: aver convertito alcuni medici obiettori. Su Repubblica di ieri, il dottor Salvatore